What is Cosine Correction？
Cosine is a mathematical function used to calculate and specify the characteristics (length & angles) of triangles (Trigonometry). Below right picture is the cosine correction comparison between MK350S Premium and Lambert’s Cosine Law. The ideal value of the red circle was described by Johann Hein¬rich Lambert in 1760 and is called “Lambert’s Cosine Law”. Basically it shows how light intensity changes as the angle of the light changes. MK350S Premium cosine receiver is the blue circle; the optimized cosine and Lambert's cosine law are almost overlapped. It is more in line with international standards for illuminance measurement - JIS AA and DIN B standards and reduced measurement errors effectively.
Cosine Correction and LUX MeasurementCosine correction is associated with measuring illuminance, or how much light brightens a certain area - it is measured in LUX which is lumens per sq. meter. A lumen is a unit of mea-sure that represents an “amount” of visible light. As you can see on the picture below, the yellow area is the total amount of the visible light which is emitted by the bulb; the visible light received in the orange 1 sq. meter area is called LUX.
LUX measurement at Different AnglesNext, you must know that light coming in at an angle will influence the LUX. A light shining from directly above (below left picture LUX: 476 lx) will shine brighter than a light coming in at an oblique angle (below right picture LUX: 449 lx). In fact there is a direct correlation between the angle of the light and amount of brightness - this was explained by Johann Heinrich Lambert in 1760.
UPRtek Spectrometer meets CIE Standard of Cosine Correction for LUX MeterMeasuring light intensity on a flat light meter lens is not so straight forward, especially when trying to measure light coming in at angles and then accurately recording the brightness according to the Lambert’s laws – you need to make adjustments. These adjustments are called Cosine Correction. The UPRtek produced spectrometer conform the cosine correction of the CIE standard for LUX Meter in terms of LUX measurement, that is to say, UPRtek spectrometer meets the characteristics of the LUX Meter. The actual LUX value of the light source (light intensity) come from different angles should be Eθ = E0 Χ cos(θ). E0 is the original LUX value. Below chart is a guide to measuring brightness according to Lambert’s laws.
Cosine Correction for Oblique Angle LightIdeally, Cosine Correction for oblique (angular) light should follow the graph shown above. For example, a light meter received lights coming in at 60 degrees; it should result in about 50% of its original intensity. As the angle becomes less oblique, the intensity should increase according the circular red line in the graph above.
- Related Products
MK350S Premium is a Handheld Spectrometer with Muli-Functions in itself. It's not only a Spectrometer, but also can be a Quantum PAR Meter, and Oscilloscope. Especially we add the BLH Judgement feature in it which's the latest popular topic and requirement for the Lighting industry. The strong and various functions can make users task quicker and efficiently.
- Related FAQ
- An undesired change in visual perception induced by a light stimulus whose luminance or spectral distribution fluctuates with time, for an observer in a certain environment. Read More
Index Defined by Definition Point Formula Index range Percent Flicker Illuminating Engineering Society (IES) A relative measure of the cyclic variation in the amplitude of a light. 0% ~ 100%
The lower the percent flicker, the less substantial the flicker.
Flicker Index Illuminating Engineering Society (IES) A measure of the cyclic variation taking into account the shape of the waveform. 0 ~ 1
The lower the flicker index, the less substantial the flicker.
SVM (Stroboscopic Effect Visibility Measure) PHILIPS & CIE Time domain → frequency domain.
Summation of weighted Fourier components
SVM < 1，Not Visible
SVM > 1，Visible
SVM = 1，Just Visible
- About Us
- Industry Category
- Measurement Case Study
Product - Handheld Spectrometer
- Is there any international certification for UPRtek Handheld Spectrometers?
- How to get the latest version firmware on my UPRtek spectrometer?
- What kind of battery should I use for MF250N?
- Why the SD card always stuck on the slot while in / out the slot?
- How long can I use UPRtek handheld spectrometer after battery had fully charged?
- Why my SD card cannot be used on UPRtek handheld spectromter?
- How to use the computer to read the measurement data of the SD card in the handheld spectrometer?
Product - uSpectrum SW
- How to download the uSpectrum PC Software?
- How to download the uFlicker PC Software?
- Is my handheld spectrometer compatible with uSpectrum PC software?
- Is my handheld spectrometer compatible with uFlicker PC software?
- How can I make the test report by uSpectrum PC software?
- Can I put my company logo on uSpectrum test report?
Product - Wing WiFi Card
- Do I need to register before I use Wing WiFi card with UPRtek handheld spectrometer?
- Does UPRtek handheld spectrometer will save the data on the same time when I process save data on smart device?
- After using uSpectrum APPs and save the data, where can I find the saved spectrum / CIE images through uSpectrum APPs on smart device?
Product - Measuring Light
- Dose UPRtek handheld spectrometer has coorected color temperature (CCT) when measuring monochromatic light?
- Could UPRtek handheld spectrometer measure the UV or IR Spectrum?
- Could UPRtek handheld spectrometer measure the light that except LED light? Such as: fluorescent lamps (fluorescent lamps) ?
- What is dark calibration? Do I need to do dark calibration every time I turn on my spectrometer?
- Knowledge - Artificial Light
- Knowledge - Light
- Knowledge - Color
- Knowledge - Tech Notes
- Product - Handheld Spectrometer
- Contact Us
UPRtek What is Cosine Correction？ Introduction
United Power Research Technology Corp. is Taiwan supplier and manufacturer in Optoelectronics Industry. UPRtek has been offering our customers high quality Spectrometer, Color Meter, Spectral PAR Meter, Spectral Color Meter, Spectrum, LED Meter, Spectradiometer, LED Measurement, LED light testing since 2010. With both advanced technology and 8 years experience, UPRtek always make sure to meet each customer's demand.